- Citado por SciELO
- Citado por Google
- Similares en SciELO
- Similares en Google
versión impresa ISSN 1678-7757
J. Appl. Oral Sci. vol.18 no.3 Bauru mayo/jun. 2010
Maria de Paula CaldasI; Flávia Maria de Moraes Ramos-PerezI; Solange Maria de AlmeidaII; Francisco Haiter-NetoIII
IDDS, MSc, PhD, Department of Oral Diagnosis, Piracicaba Dental School, State University of Campinas, Piracicaba, SP, Brazil
IIDDS, MSc, PhD, Associate Professor, Department of Oral Diagnosis, Piracicaba Dental School, State University of Campinas, Piracicaba, SP, Brazil
IIIDDS, MSc, PhD, Full Professor, Department of Oral Diagnosis, Piracicaba Dental School, State University of Campinas, Piracicaba, SP, Brazil
OBJECTIVE: The aim of this study was to establish which materials afford better simulation of soft tissues in Oral Radiology studies.
MATERIAL AND METHODS: The sample was composed of four materials in eleven different thicknesses to simulate the soft tissues of the face. The mean values of the relative amounts of radiographic contrast of the materials were determined and compared to a gold standard value, which was obtained from 20 patients who were referred to have periapical radiographs taken of the left mandibular molars. Data were subjected to statistical analysis with Dunnett's test (p<0.05).
RESULTS: The mean value of the relative amounts of contrast encountered in the patients was 0.47, with a range between 0.36 and 0.64 for all 44 material/thickness combinations. The majority of the tested materials showed values close to those of the patients' tissues, without statistically signifcant differences among them. The values of only three materials/thickness combinations differed statistically from those of the patients' tissues.
CONCLUSIONS: Based on the results of the present study, it may be concluded that except for utility wax (4 mm and 8 mm) and water (4 mm), all materials tested at different thickness could be used as soft tissue substitute materials in Oral Radiology studies.
Key words: Tissues. Radiography dental. Simulate.
In Oral Radiology research, phantoms are frequently used to simulate the patient's body. There are, however, some requirements for tissue substitutes. A material that could be easily obtained and simulate the soft tissue would be helpful for Oral Radiology professors and researchers. It is important for these materials to be capable of being accurately measured, available, reproducible, and ready to be used in any instance7. Thus, standardization would be perfectly possible. The development of phantoms that present densities similar to those observed in patients is very important because it avoids unnecessary radiation exposure. According to the ALARA principle, all unnecessary exposure to radiation should be avoided15.
Dry mandibles are widely used in optical bone density studies3,5,6,10. However, the studies previously reported in the literature do not consider the signifcant infuence of the patient's soft tissue located between the bone and the x-ray beam. The intensity of an x-ray beam is reduced by interaction with the matter it encounters. This attenuation results from interactions of individual photons in the beam with atoms in the absorber. The x-ray photons are either absorbed or scattered out of the beam15. It is expected that this phenomenon also occurs when the x-ray beam interacts with the mandible. It is thus very important to consider the presence of a specifc material that simulates the soft tissue of the human face when obtaining the optical value of the jaw density 3.
Various materials simulating soft tissues have been cited: water, wax, self-polymerizing resin, paraffin and polyethylene1-3,5,9,13. However, all studies consider bovine muscle as gold standard to test a specifc soft tissue substitute material.
A specifc substitute tissue that presents contrast values similar to those of humans is essential and might facilitate all radiological experiments and education. Thus, the aim of this study was to establish which materials offer better simulation of soft tissues in Oral Radiology studies, using the human soft tissue as gold standard.
MATERIAL AND METHODS
The sample was composed of four different materials in eleven different thickness (4, 8, 12, 15, 20, 24, 28, 32, 36, 40 and 45 mm): self-polymerizing acrylic resin, utility wax, wood and a 2 mm-thick polymethylmethacrylate box filled with a 2-mm-thick water layer. All materials were exposed using InSight dental film (Eastman Kodak Co., Rochester, NY, USA) with the addition of an aluminum step-wedge. Standardized conditions were used: GE 1000 machine (General Electric Co., Milwaukee, WI, USA), operating at 70 kVp, 10 mA, and 40 cm focus-film distance. The dental films were attached to a dry mandible, with buccal interposition of a soft tissue substitute material, and the x-ray beam was projected perpendicular to the film (Figure 1). Three standardized radiographs of each material at its specifc thickness were obtained from the mandibular posterior segment of a cadaver.
Using a densitometer (MRA, Ribeirão Preto, SP, Brazil), it was possible to determine the radiographic density value of all materials at their specifc thicknesses. This value was used to obtain the relative amounts of contrast or contrast index, according to Price12 (1986), using the following equation:
C= D2-D7/ 0.5(D2+ D7), where C is the relative amounts of contrast or contrast index, D2and D7 are the second and seventh step of the density scale, respectively.
Radiographs from 20 patients referred to have periapical radiographs of mandibular left molars taken at the Oral Radiology Clinic of Piracicaba Dental School, State University of Campinas, were used. The research protocol was approved by the Research Ethics Committee of Piracicaba Dental School, and the patients signed an informed consent form before their enrollment.
All radiographs were taken in accordance with routine procedures. The patients wore lead aprons and leaded thyroid collars in order to enhance the diagnostic benefts of dental radiographs and minimize patient's exposure to radiation. All images were exposed with the same GE 1000 machine, InSight dental film and focus-film distance. Using a densitometer, it was possible to determine the radiographic density value of each patient. This value was used to obtain the soft tissue relative amounts of contrast or contrast index, according to Price12 (1986), as mentioned above. The mean of these values was used as gold standard.
All films were processed using a Gendex GXP (Gendex Dental Systems, Lake Zurich, IL, USA) with fresh Kodak processing liquids and operating time of 5 min.
The mean values of the relative amounts of radiographic contrast of each material (at its specifc thickness) were compared with the gold standard value using Dunnett's statistical analysis (p<0.05).
The results showed that the mean value of the relative amounts of contrast encountered in the patients was 0.47, with a range between 0.36 and 0.64 for all 44 material/thickness combinations. The majority of the tested materials showed values close to those of the patients' tissues, without statistically signifcant differences among them. It was possible to observe that the values of only three materials/ thickness combinations differed statistically from those of the patients' tissues, utility wax (4 and 8 mm) and water (4 mm), which presented mean relative amounts of contrast of 0.64, 0.63 and 0.62, respectively (Table 1).
When analyzing the four different materials at each specifc thickness (4, 8, 12, 15, 20, 24, 28, 32, 36, 40, 45 mm), there was no statistically signifcant difference among them. However, when comparing one specific material at its eleven different thicknesses, there were statistically signifcant differences among some of the thicknesses (Table 1).
As it is impossible to measure doses within the patient, many tissue-equivalent materials have been developed, and dosimetric studies have been conducted in phantoms that approximate the human form to refect the x-ray absorption and scattering properties of various tissues4. The materials at the specifc thicknesses evaluated in the present study are ideally appropriate for radiographic, dosimetric and radiobiological studies.
Water was the first soft tissue substitute material to be used in radiation measures and up to now it continues to be tested5. Blake, et al.1 (1992) studied the effects of beam hardening on measurements made with a commercial dual energy x-ray scanner. Bone was represented by layers of aluminum of linearly increasing thickness, which were scanned under water thicknesses ranging from 0 to 25 mm to represent different body thicknesses of soft tissue. Borg, et al.2 (1998) placed jaw specimens immediately behind a polymethylmethacrylate cylinder filled with 20 mm of water to simulate the soft tissue of the face. In the present study, a 2-mm-thick polymethylmethacrylate box filled with water was used. No signifcant difference was found between 8, 12, 15, 20, 24, 28, 32, 36, 40, 45 mm of water and the patient's tissues.
Meurer11 (2003) testing samples of different soft tissue substitute materials to study the optical values of the human jaw, advocated that 20 mm of acrylic was the material that best reproduced the results found with the muscular tissue, used as gold standard. Gegler, et al.8 (2006) joined a 20-mm- thick acrylic block to a maxilla simulator model in order to simulate the soft tissue of the face. The acrylic used in this research ranged from 4 to 45 mm thick. However, none of the thicknesses tested presented statistical difference when compared with the patient, indicating that 4, 8, 12, 15, 20, 24, 28, 32, 36, 40 and 45 mm of acrylic can be used as soft tissue substitute of the face.
Brand, et al.4 (1989) constructed a phantom to obtain accurate estimates of radiation doses in the head and neck region. The soft tissues of the head and neck were represented by a mixture of wax, plastic, magnesium oxide, and titanium dioxide with the x-ray absorption and scattering properties close to those of water and soft tissue. Soft tissue thicknesses were based on depths reported in the literature and supplemented by cadaver measurements. Conversely, the soft-tissueequivalent material used in this study was utility wax alone, due to its availability and being easy to use. The wax thickness ranged from 4 to 45 mm, and only two values differed statistically from those of the patient's tissues: utility wax 4 mm and 8 mm, which presented mean relative amounts of contrast of 0.64 and 0.63, respectively.
Demann, et al.6 (2002) investigated the effects of soft tissue and position on vector during distraction. The authors used polyethylene straps in the temporomandibular joint region in a manner that resembled the origin and insertion of the masticatory muscles. They concluded that simulated soft tissues of the face affected the vector of distraction. Other materials, such as epoxy resin and hydrophilic materials, have also been used to substitute soft tissue in Oral Radiology7,14.
In this study, the materials were chosen according to their availability and reproducibility. The thicknesses of the materials were determined based on the reference values of the patients. Thus eleven different thicknesses were established for each material in order to find out the specifc thicknesses that presented no statistical difference from those of the patients. The contrast values of the materials tested resemble human biological tissue.
Phantom materials are used to simulate the interactions of electromagnetic radiation with the body tissue and organs. A material that scatters and absorbs radiation in a similar way as that of the body is a potentially useful phantom material7. However, there are very few reports in the literature concerning materials used as soft tissue substitutes in Oral Radiology research. In this study, different possibilities of materials and thicknesses that can be used to replace soft tissues in Dentistry have been presented.
The clinical signifcance of the present work is related to the fact that the reference values were obtained from patients with different tissue densities. Therefore, the patient's muscle used as gold standard was more precise and reliable.
It was possible to conclude that except for utility wax (4 mm and 8 mm) and water (4 mm), all materials tested at different thickness could be used to simulate soft tissues in Oral Radiology studies.
1- Blake GM, McKeeney DB, Chaya SC, Ryan PJ, Fogelman I. Dual energy x-ray absorptiometry: the effects of beam hardening on bone density measurements. Med Phys. 1992;19:459-65. [ Links ]
2- Borg E, Kallqvist A, Grondahl K, Grondahl HG. Film and digital radiography for detection of simulated root resorption cavities. Oral Surg Oral Med Oral Pathol Oral Radiol Endod. 1998;86:110-4. [ Links ]
3- Braga CPA, Gegler A, Fontanella V. Avaliaçao da infuencia da espessura e da posição relativa de materiais simuladores de tecidos moles na densidade optica de radiografas periapicais da região posterior da mandíbula. Cienc Odontol Bras. 2006;9:52-8. [ Links ]
4- Brand JW, Kuba RK, Braunreiter TC. An improved head-and-neck phantom for radiation dosimetry. Oral Surg Oral Med Oral Pathol. 1989;67:338-46. [ Links ]
5- Cook JE, Cunningham JL. The assessment of fracture healing using dual x-ray absorptiometry: a feasibility study using phantoms. Phys Med Biol. 1995;40:119-36. [ Links ]
6- Demann ET, Haug RH. Do position and soft tissue affect distraction vector? An in vitro investigation. J Oral Maxillofac Surg. 2002;60:149-55. [ Links ]
7- Farquharson MJ, Spyrou NM, al-Bahri J, Highgate DJ. Low energy photon attenuation measurements of hydrophilic materials for tissue equivalent phantoms. Appl Radiat Isot. 1995;46:783-90. [ Links ]
8- Gegler A, Mahl C, Fontanella V. Reproducibility of and file format effect on digital subtraction radiography of simulated external root resorptions. Dentomaxillofac Radiol. 2006;35:10-3. [ Links ]
9- Hildebolt CF, Rupich RC, Vannier MW, Zerbolio DJ Jr, Shrout MK, Cohen S, et al. Inter-relationships between bone mineral content measures. Dual energy radiography (DER) and bitewing radiographs (BWX). J Clin Periodontol. 1993;20:739-45. [ Links ]
10- Lindsay DD, Stern BE. A new tissue-like material for use as bolus. Radiology. 1953;60:355-62. [ Links ]
11- Meuer MI, Meuer E, Yurgel LS, Costa NP. Análise da densidade óssea em região parassinfisária de mandíbulas humanas: comparaçao entre niveis de cinza em radiografas digitais (sistema digora) e unidades hounsfield. Rev Odont Cienc. 2003;18:179-86. [ Links ]
12- Price C. The effects of beam quality and optical density on image quality in dental radiography. Oral Surg Oral Med Oral Pathol. 1986;62:580-8. [ Links ]
13- Richards AG, Webber RL. Constructing phantom heads for radiation research. Oral Sur Oral Med Oral Pathol. 1963;16:683-90. [ Links ]
14- White DR, Martin RJ, Darlison R. Epoxy resin based tissue substitutes. Br J Radiol. 1977;50:814-21. [ Links ]
15- White SC, Pharoah MJ. Oral Radiology. Principles and Interpretation. St. Louis: Mosby; 2000. [ Links ]
Maria de Paula Caldas
Av. Limeira, 901
13414-903 - Piracicaba, SP - Brasil
Phone: +55 (19) 2106-5327 - Fax-number: +55 (19) 2106-5327
Received: December 12, 2008
Modifcation: July 19, 2009
Accepted: September 16, 2009